Technical component Modification of the Testomat® 808 & 808SIO ₂
Dear customers and business partners,
Dear service technicians,
We would like to inform you about a technical change of the
Pump head / double pump head from Gebrüder Heyl Analysentechnik
GmbH & Co.KG for the Testomat 808 and the Testomat 808 SiO₂ inform.
Please refer to the enclosed "Information flyers" the procedure
You can find the instructions for replacing the pump head / double pump head
in the download area under General Product Information Flyer or in the link below.
The new pump heads are fitted as standard from device number 253.061.
As a spare part, the new pump head Testomat 808 under the item no. 37562 and for
the Testomat 808 SiO2 under the Art.-NR. To acquire 37859.
For replacing the pump head for the Testomat 808 up to device number
252.776 you need the new conversion kit 808 pump head of the art. No.
40363 is to order.
For the Testomat 808 SiO2 you have the new device number 251.610
Conversion kit double pump head for Testomat 808 SiO2 now available.
Here there will be no price increase over the previous products
with the item no. Give 037578 & 037801!
The prices remain the same according to the current price list from 01.04.2019.
We refer to the current conditions of sale as of 25.06.2019.
If you have questions, suggestions or hints, please do not hesitate to contact us
at any time by phone or by e-mail a message below
+49 (0) 5121 / 7609-0 or email@example.com.
Technical information flyer pump head / double pump head
We thank you for the implementation and your trust in us!
Gebr. Heyl distribution company for innovative water treatment mbH